I aligned X arm so that the beam spot comes roughly on the center.
1. Use ITMX and ETMX (mainly ITMX) to make beam spot come on center of the optic using eyeball.
2. Use ETMX and BS to maximize TRX power (reached ~ 0.85)
3. Aligned green optics on X end. Transmission of X green measured at PSL table is now 255 uW and TEM00 has the most power.
It was not easy to increase X green transmission more because beam spot on green transmission PD is wiggly when X end table is opened. When closed, wiggliness is about the same for Y green and X green.
Turning off HEPA on the X end didin't helped, but there must be something bad in the X end table. If we couldn't figure out why, let's wait for PZTs to come for end tables.
Considering the laser power is different(X end 1 W, Y end 700 mW), X green transmission should reach ~400 uW. But I think we should go on to X beat search.
I placed green shutter for X end back for convenience. I put some spacers to adjust its height and avoid beam clipping.
What causing wiggly X green transmission was the air flow from the air conditioner. When we turned it off, beam spot motion became quiet. Air flow from HEPA was not effecting much.