When I've put Guralps inside the isolation box, the signal from seismometers increased and was out of AA board range. I've reduced the gain of the readout box by a factor of 2. Now R2 for channels 1-6 is (2000, 1050, 1050, 2000, 1050, 1050) Ohm.
The signal increased in the frequency range 30-50 Hz. Guralp noise become better. That's good. However, it is still worse then in the manual.
As Yuta is dancing on the isolation box, Guralp signal is most time out of the AA board range. So I calculated the noise based on 5 min data. This may be enough, but I'll repeat the experiment later with 30 min data.