The following tasks need to be done in the daytime tomorrow.
For ITMX, I used the values from the conlog:
2011/08/12,20:10:12 utc 'C1:SUS[-_]ITMX[-_]INMATRIX'
These are the latest values in the conlog that aren't the basic matricies. Even though we did a round of diagonalization in Sept, and the
matricies are saved in a .mat file, it doesn't look like we used the ITMX matrix from that time.
For PRM, I used the matricies that were saved in InputMatricies_16Sept2011.mat, in the peakFit folder, since I couldn't find anything in the Conlog other than the basic matricies.
UPDATE: I didn't actually count the number of oscillations until the optics were damped, so I don't have an actual number for the Q, but I feel good about the damping, after having kicked POS of both ITMX and PRM and watching the sensors.