I've measured the input signal to the seismic box from seismometer Guralp 1. The spectrum of the signal in the "input +" (TP 1) is
The spectrum below 1 Hz is ~250 uV/sqrt(Hz). As the input is differential, then the input voltage is 0.5 mV/sqrt(Hz). The spectrum of the "output +" signal (TP 2) is
So the gain at ~ 1Hz is ~20. I've measured the transfer function between the "input +" and "output +" (TP1 and TP2) for all 9 circuits
The channels 1-6 are of new modification and have gain ~20 at the frequencies 0.2 - 100 Hz. Below 0.2 Hz the gain is reduced. 100 Hz - cut off frequency of the low-pass filters. Meanwhile channels 7-9 (old configuration) have much more gain and 10_50 Hz filters work here.
The coherence between "input +" and "output +" (TP1 and TP2) for 9 circuits is
We can see that channel VERT 3 is very bad. For others coherence is lost below 0.2 Hz. The spectrum analyzer noise measured is ~1000 times less then the signal at these frequencies. I'll pay more attention to this loss of coherence at low frequencies. Something is noisy.