[John / Valera / Kiwamu]
We installed a new weapon, an optical spectrum analyzer in the AS port.
Like we used to do in the old days, two BNC cables were newly laid down and they bring the output of the OSA to the control room to monitor the spectrum with an oscilloscope.
The photo diode of the OSA was replaced by a Thorlab PDA100A to amplify the signals.
The carrier peak is at about 6.9 V and the f1 and f2 sidebands peaks are at about 40 mV when the beam is in straight shot (everything is misaligned except ITMY and BS).
According to a rough calculation, those numbers correspond to a modulation depth of about 0.16 or so.
The depth agree with what Mirko measured before (#5519)