I have been looking at the swings in the RAMmon channels since the heater was reengaged, to compare them to the data from beforehand (with and without the foam box). With the large grains of salt that I will list after, it appears that the EOM temperature controller does in fact reduce the amplitude of the swings by a measurable factor.
It seems impractical to try and rope off essentially 3 straight days where nothign major can be done to the IFO just to take RAM data. Instead, I think we should figure out a way to mimic the diurnal temperature swings on ~hour timescales. The EOM can temperature follows PSL-FSS_RMTEMP almost exactly and with a very short delay, so we can probably even accomplish this by stepping the lab A/C temperature. If this won't work, we can use an incandescent lamp or something similar to heat up the area around the EOM by a noticeable amount.
I'll try to come up with a good way to do this so that we can get some reliable data...