Not logged in
Mon Jun 30 18:36:01 2008
MZ alignment again
We re-adjusted the MZ alignment. The reason behind this is to make sure that the MZ dark port is not falling at a strange fringe, where it is only dark at the dark port PD. It can happen when the two beams poorly overlap.
We tried both the minimization of the MZ dark PD and the maximization of the MZ transmission at the same time.
We also placed another PD in the MZ dark port at a different distance from the original dark PD and tried to minimize this too.
If the MZ dark port is at a strange fringe, one of the dark PD can be dark where the other one is still bright.
If both of the dark PD get dark, the overlap between the beams should be ok.
We tweaked only the two mirrors of the MZ after the EOMs (mainly the one with a PZT).
Right now, the MZ dark power is 0.432.
BTW, we should change the name of the MZ dark port on the medm screen (it is now MZ reflection, where it is not a reflection).
I will try to change it later.
We wanted to put the beam position on the IOO-QPD_POS_* back to the original (before John tweaked the MZ alignment earlier).
However, the trends of IOO-QPD_POS_* show a lot of fluctuation and jumps, of which we don't know the cause. So we could not find reasonable original values.
We suspect a circuit problem in IOO-QPD_POS, especially because the jumps are very strange.
We will do this investigation later too.