Here is a 48 hours trend of the RFAM monitor (a.k.a StochMon):
The upper plot is the DC output from the StochMon PD and the lower plot shows the calibrated RIN (Relative Intensity) at each modulation frequency.
I have downloaded minutes trends of StochMon for 48 hours staring from 6:00AM of Nov/24.
I followed Koji's calibration formula (#6009) to get the actual peak value (half of the peak-peak value) of the RF outputs and then divided them by the DC output to make them RIN.
It looks the RINs are hovering at ~ 4 x 10-4 and fluctuate from 1x10-4 to 1x10-3. Those numbers agree with what we saw before (#5616)
So it seems the StochMon is working fine.
New RFAM mon calibration