Some more info on this:
At these frequencies the pendulum should be quieter than the stacks. By quite a bit actually since there is the stack resonance at a couple Hz. 'Glueing' them together via the local control is not wise. We put an elliptic LP ( 2.5Hz, 4th order 6dB) into the C1:SUS-MC?_SUSPOS pathes and MC-F got better above 1Hz
Added an extra LP @ 10 Hz afterwards. Doesn't make a visible difference.
Now here is more stuff to consider.
1. The OSEMs glue the MC mirrors to the stacks
2. The pendulum TF should be 1
3. It shouldn't matter if the OSEMs do or do not act on the mirror at these frequencies, assuming they don't add extra noise.
4. Page http://nodus.ligo.caltech.edu:8080/40m/5547 seems to indicate OSEM sensor noise is so low it can be neglected.
If we reduce the gain in the OSEMs by adding additional HP filters ( cheby2, HP 0.3Hz, 6dB 4th order ) the happens:
1. MC length gets a bit more noisy at low frequencies - should be looked into some more
2. Coherence between the GUR1 seismometer and MC length goes up between 1E-2 and 1E-1 Hz:
( Ref is with low OSEM gain )
The stacks might be more correlatedly moving together than the pendulums. This would be not so nice for OAF test, but really fine for actually using the MC.
Todo: Measure the OSEM to seismometer coherences with high and low OSEM gains.
For reference the seismometer coherence with one another: