Not logged in
Thu Jun 26 21:07:11 2008
dust & MC1
I realized today, that yesterday while we were trying to debug the adaptive noise canceler, I turned
off the analog dewhitening on MC1. I did this by changing settings on the Xycom screen but I
forgot to elog this -- this may have caused problems with locking via increased frequency noise.
I have now returned it to its nominal, dewhitening on, configuration.
I also used mDV to look at the last year of dust trend. I have plotted here the cumulative
histogram in percentile units of the 0.5 micron dust level. The x-axis is in units of particles per cu. ft.
and the y-axis is percentage. For example, the plot tells us that over the last year, the counts were
below 6000, 90% of the time. I have set the yellow and red alarm levels to alarm at the 95-th and 99-th
percentile levels, respectively.