Not logged in
Wed Jun 25 18:03:21 2008
FSS Input Offset slider problem
While working on the PMC scanning, I noticed that the FSS input offset slider is doing nothing.
I traced the signal flow and checked the cables/boards.
The slider changes the output voltage from a VMIVME4116 DAC in the PSL rack. This output voltage is confirmed to be correct at the FLKM64 connector. The signal is connected to the FSS servo interface box (D040423) trough a ribbon cable. However, the output from the interface box is always -27V regardless of the slider position.
Therefore, either the interface box (D040423) or the ribbon cable has a problem.
I will debug the interface box using an extension card when no one is working on the interferometer.