The input matrices of ETMX, ITMX and PRM have been newly inverted.
Those were the ones having some troubles (see #5444, #5547).
After a coarse adjustment of the damping Q factors, they look damping happily.
Before running the peakFit scripts, I woke up the nds2 sever process on Mafalda because it hasn't been recovered from the power outage.
To start the nds2 process I followed the instruction by Jamie (#5094).
Then I started requesting the data of the last night's free swinging test (#5594)
However the NDS2_GetData command failed everytime when data with long duration were requested.
It maybe because some of the data are missing in sometimes, but I haven't seriously checked the data stored in fb.
So for the reason, I had to use a short duration of 1200 sec (default is 3600 sec). That's why spectra look noisier than usual.