Not logged in
Tue Jun 24 17:12:10 2008
GC750 setup, 1X4 Hub connected, ETMX images
The GC750 camera has been setup to look at ETMX. In addition, the new 1X4 rack mounted switch (220.127.116.11) has been connected via new cat6 cable to the control room hub (18.104.22.168?), thanks to Eric. The camera is now plugged into 1X4 rack switch and now has a gigabit connection to the control room computers as well as Mafalda (22.214.171.124).
By using ssh -X mafalda or ssh -X 126.96.36.199, then typing:
A viewer will be brought up. By clicking on the 3rd icon from the left (looks like an eye) will bring up a live view.
Closing the view, and then cd ../../40mCode, and then running ./Snap --help will tell you how to use a simple code for taking .tiff images as well as setting things such as exposure length and size of image (in pixels) to send.
When the interferometer was set to an X-arm only configuration, we took two series of 200 images each, with two different exposure lengths.
Attached are three pdf images. The first is just a black and white single image, the second is an average of 100 images, and the third is the standard deviation of the 100 images.