Not logged in
Tue Jun 24 10:24:43 2008
Abs. Len. Meas. ~ Cavity Swing Measurement (2)
At the entry 555, Alberto reported the results of the cavity length measurement using cavity sweeping.
As expected, each result inevitably has an ambiguity depending on which resonance do we take as an upper sideband.
In order to exclude this ambiguity Steve and Koji performed a primitive non-optical measurement using a tape and photos:
This morning Steve and Koji did tape measurements to know the lengths between the ITM/ETM chambers.
Yesterday, Koji took photos of the optical tables in vacuum to know the actual positions of the suspensions.
The results are shown in the figures attached. From those non-optical measurements the lengths of the X/Y arm are known to be 38.48+-0.03 / 38.67+-/0.03 [m].
Then, we could exclude the shorter lengths of the values in the entry 555. i.e. The Y arm is longer than the X arm about 0.2 m.
These approximate lengths will be used in the further precise measurements which use precise scans of the FSR frequencies.