As a suspension test I am leaving all of the suspensions restored and damped with OSEMS but without oplevs
According to the spectra, all of the suspensions had been damped with the OSEMs. The peaks around 1Hz are reasonably suppressed.
However the spectra from ITMX showed a noise floor at very high level. This is because of strange jumps in the signal of the UL shadow sensor.
I will check some analog circuits for the UL sensor.
(ITMX shadow sensors)
Here is the spectra of the ITMX shadow sensors taken during the damping test (#5534)- -
The UL sensor shows a unacceptable amount of noise.
Additionally I checked the time series of the ITMX shadow sensors and found ONLY the UL sensor frequently showed strange jumps in data.
Here is an example of the time series showing a jump ONLY in the UL sensor.
It is possible that the jumps are coming from some circuits, since the rest of the sensors (including the oplevs) don't detect the same jump.