Both loops basically have no phase margins. i.e. unstable. How can you lock PRMI with these servos?
The following rough swept sine plots are the OLTs for MICH and PRCL. The gain setting was -10 and 0.5 for MICH and PRCL, respectively. Integrators were off. Looking at the measured plots, MICH has about 300 Hz UGF, when the gain is -20, and PRCL has about 300 HZ UGF, too, when the gain is 0.8.