Another Hamasutu S3399 photodiode was tested with the electronic circuit as described in LIGO-D-1002969-v.
RF transimpedance is 1k although the DC transimpedance is 2k.
The noise level is 25pA/sqrt(Hz) which corresponds to a dark current of 1.9mA or 1.7mA in the independent measurement.
At all frequencies the noise is larger compared to Koji's measurement (see labbook page 4778).
In file idet_S3399.pdf the first point is not within its error bars on the fitted curved. This point corresponds to the dark noise measurement
I made this measurement again. Now it is on the fitted curve. In the previous measurement I pushed the save button a bit too early. The
averaging process has not been ready while I pushed the 'save' button.
Dark current is 1.05mA and noise is lower than in the previous measurement.
New file are the XXX_v2.pdf files