New channels, POP55 and POY11 are connected to the rack and now available on the data system.
POX11 I is not working. I didn't investigate what was wrong. Please make sure when you come to need POX11.
The orthogonalities of POY11 and POP55 were measured and already adjusted. The results are below:
ABS = 0.973633
PHASE = 92.086483 [deg]
ezcawrite C1:LSC-POY11_Q_GAIN 1.027081 && ezcawrite C1:LSC-POY11_PHASE_D 92.086483
ABS = 1.02680579
PHASE = 88.5246 [deg]
ezcawrite C1:LSC-POP55_Q_GAIN 0.973894 && ezcawrite C1:LSC-POP55_PHASE_D 88.524609