The new ITMX was aligned by changing the DC biases.
The resultant DC biases are reasonably small.
C1:SUS-ITMX_PIT_COMM = -0.2909
C1:SUS-ITMX_YAW_COMM = -0.0617
The alignment was done by trying to resonate the green light in the X arm cavity.
The spot position of the green light on the ITMX mirror looked good. This was confirmed by inserting a sensor card.
I did the OSEM mid-range adjustment and the rotation adjustment but at the end the OSEM DC voltage has changed due to the DC bias operation.
The OSEM rotation was approximately optimized so that all the face shadow sensors are sensitive to the POS motion but the SIDE shadow sensor is insensitive to the POS motion.
It needs a free swinging diagnosis.
ITMX OSEMs UL 1.8V, UR 1.7V, LR 0V, LL 0V, SD 1.3V at the same bias setting shown above. May be a lose earth quake tip?or magnet is touching?