Not logged in
Fri Jun 13 11:17:25 2008
PMC transmittance at high power
We received a new cable for the Scientech calorimeter. So I measured the transmittance of the PMC at higher power.
Input power = 2.298W
Output power = 1.364W
Transmittance = 59%
The input power to the PMC was measured between the two mode matching lenses by the calorimeter.
2.298W looks a bit too low. Actually, the calibrated monitor PD on the MEDM screen shows about 3W output from MOPA.
So we (me and Steve) measured the power right after the PBS after the periscope from MOPA with the HWP set to maximize the transmission of the PBS.
It was 2.77W. According to Steve's previous measurement, the first mirror of the periscope transmits about 200mW of the incoming light to the monitor PD. So the actual output of the MOPA is about 2.97W, which is consistent with the monitor PD reading.
The aperture of the EOM for the PMC control is glowing a lot. We suspect this is the main cause of the loss (from 2.77W to 2.298W).
We may want to re-align the EOM.
The output light from the PMC was picked off by a glass slide. The reflectance of the glass slide was measured first at a lower power (input 98mW, reflected power 1.58mW). Assuming that the reflectance is the same for the higher power, I turned up the input power to the PMC. This time, the picked off power was 22.45mW. This means the actual output power is 98/1.58*22.45=1364mW. The glass slide was kept at the same angle through out the measurement.
The measurement of the output power was done by the Ophir power meter. So calibration difference between the Ophir and the calorimeter may introduce some error.