After the IFO was aligned in air one final time, we tapped on a few OSEMs until we were happy with all of the centering of all of the optics' OSEMs. All are within 0.05 of their halfway values, with the exception of one each on MC1 and MC3, one of which is within 0.06, and the other 0.08. Because of the realignment pain of dealing with MC OSEMs, we elected to leave these alone. Also, since we obviously didn't open the MC2 tank, we don't know how they are, although the numbers look reasonable.
Also, we took photos (to be posted on Picasa in a day or two) of all the main IFO magnet-in-OSEM centering, as best we could. SRM, BS, PRM all caused trouble, due to their tight optical layouts. We got what we could. Various people have been looking at these for the past 2 weeks, and I think they're all fine, even if we didn't get stellar photos.
We are now prepared for pumping. For real this time.