[Kieko / Kiwamu]
The AS beam shows a little bit of clipping and right now this is the only concern for the alignment of the interferometer.
Other than that everything is okay including :
+ Beam centering on all the suspended optics
+ Arm cavity alignments. The Fabry-Perot fringes from both arm cavity were found on the AS camera.
+ POX/POY/POP, they are still successfully coming out from the chambers
+ Alignment of the green beams and the associated optics. Both green are reaching to the PSL table
We need to check/fix the AS beam clipping and once it's done we will readjust the OSEM mid range and the oplevs.
Then it will be ready for the drag/wipe and door closing.