I have re-calibrated the photosensor I used to measure the displacements of the TT frame (what I call "Photosensor 2").
As before, the linear region is about 15.2mm to 25.4mm. It is characterized by the slope -0.0996 V/mm (-0.1 V/mm). Recall that photosensor 1 (used to measure mirror displacements) has a calibration slope of -3.2V/mm. The ratio of the two slopes (3.2/0.1 = 32). We should thus expect the DC coupling level to be 32? This is not what we have for the DC coupling levels in our data (2.5 for flexibly-supported, fully-assembled TT (with EDC, with bar), 4.2 for EDC without bar, 3.2 for rigid EDC without bar, 3.2 for no EDC, with bar, 3.2 for no EDC without bar) . I think I may need to do my calibration plot for the photosensor at the frame?