I ended up choosing a different dither frequency for driving the NPRO PZT: 230 kHz, because the phase modulation response in that region is higher according to other data taken on an NPRO laser (see this entry). At 230 there is a dip in the AM response of the PZT.
I am driving the PZT at 230 kHz and 13 dBm using a function generator. I am then monitoring the RF output of a PD that is detecting light reflected off the cavity. (The dither frequency was below the RF cutoff frequency of the PD, but it was appearing in the "DC output", so I am actually taking the "DC output" of the PD, which has my RF signal in it, blocking the real DC part of it with a DC block, and then mixing the signal with the 230kHz sine wave being sent to the PZT.
I am monitoring the mixer output on an oscilloscope, as well as the transmission through the cavity. I am sweeping the laser temperature using a lock in as a function generator sending out a sine wave at 0.2 V and 5 mHz. When there is a peak in the transmission, the error signal coming from the mixer passes through zero.
My next step is to find or build a low pass filter with a pole somewhere less than 100 kHz to cut out the unwanted higher frequency signal so that I have a demodulated error signal that I can use to lock the laser to the cavity.