The PSL temperature box has returned to service, with some circuit modifications. The 1k resistors on all the temp. sensor inputs (R3, R4, R7, R8, R12, R12) were changed to 0 Ohm. Also, the 10k resistors R26, R28, R29, and R30 were changed to 10.2k metal film. The DCC document will be updated shortly. There is now an offset in the MINCOMEAS channel compared to the others, which will be corrected in the morning after looking at the overnight trend.
Forgot to do this in May. Have just changed the values in the psl.db file now as well as updating them live via Probe.
To make the appropriate change, I took the measured offset (5.31 deg) and added 2x this to the EGUF and EGUL field for the MINCO_MEAS channel. (see instructions here)
Committed the .db file to the SVN.
attached plot shows 8 days of trend with 5.31 degC added to the black trace using the XMGRACE Data Set Transformations