[Kiwamu / Steve]
We checked some electronics noise on the ETMY shadow sensor system.
Noise from the WF, AA board and ADC are below the shadow sensor spectra on ETMY.
It means something funny is going on in the upstream side (including the satelight box and shadow sensors)
OR the coil drivers side are going crazy ??
As Rana pointed out in his entry (#5025), the spectra of the shadow sensors on ETMY were quite bad below 3 Hz. The floor are higher than that of ETMX by factor of 10 or so.
To check if the noise comes from some of the electronics, we disconnected D15-sub from pd to whitening in.
The spectra with/without shadow sensors are attached below.
The curves in brown and green are the ones taken when the shadow sensors were disconnected from the WF board.
So these two curves represent the summed noise of the WF, AA and ADC.
This tells us to look toward the OSEM.