We laid the cable along the cable keeper from the BACARDI seismometer to the rack 1X6, the excess cable has been coiled under the X arm.
We plugged the cable to the seismometer and to the seismometer electronics box in rack 1X6. We also plugged the AC power cable from the box to an outlet in rack 1X7 (because the 1X6 outlets are full)
With the help of a function generator we tested the following labeled channels of AA board...
2, 3, 11, 12, 14, 15, 16, 18, 19, 20 and 24
that are the channels that can be viewed by the dataviewer, also the channel 10 can be viewed but it's labeld BAD so we cannot use it.
We leveled the seismometer and unlocked it, and saw his X,Y,Z velocity signals with an oscilloscope.