Not logged in
Tue May 20 18:33:01 2008
Mode Cleaner is locked again
It was noticed by Mr.Adhikari earlier today that the MC became unlocked at about 11AM.
There is no clear understanding what caused the problem.
Trying to restore the modecleaner locking, we noticed with John that the beam was not centered on the wavesensors (WFS1. WFS2 on the screen "C1IOO_LockMC.adl"). We decided to adjust the beam position moving slightly the bias sliders for pitch and yaw degrees of freedom for MC1.
This allowed to make the MC locked.
Old positions for the MC1 sliders: Pitch = 2.9934, Yaw = -0.6168;
New positions --------//---------: Pitch = 3.0604, Yaw = -0.7258.
At the same time, FSS for PSL is still showing the values in the range 0.720 - 0.750 which is lower than the usual values. The indicator for FSS value is yellow when it is below 0.750.