The PRM sus damping restored. C1:SUS-PRM_SDPD_VAR is still 20-30mV and going up. Side gain turned on. This pulled it down to 5-8 mV
Why is the side osem sensing voltage 4.4V ? It can not be higher than ~2.4V.......something is rotten in the state of Denmark?
Edit by KI:
It's because Valera increased the transimpedance gain of the PRM SIDE OSEM to match the signal level to the new ADC range (#3913 ).