I added a new ADC channel for a DC signal from the X end green PD.
It is called C1:GCX-REFL_DC and connected to adc_0_1, which is the second channel of ADC_0.
By the way, when I tried connecting it to an ADC I found that most of the channels on the AA board on 1X9 were not working.
Since the outputs form the board are too small the circuits may have benn broken. See the picture below.
In addition to that I realized that the signal from the PDH box for the temperature actuation is limited by +/- 2V due to the range of this AA board.
In fact the signal is frequently saturated due to this small voltage range.
We have to enlarge the range of this AA board like Valera did before for the suspensions (see this entry).