That's some pretty fast work! I thought we would be taking up to a week to get that happening. I wonder what's the right way to measure the inherent frequency noise of this thing?
Also, should the comparator part have some hysteresis (ala Schmidt trigger) or is it best to just let it twirl as is? Is it sensitive to DC offsets on the input or is there a high pass filter? What's the correct low pass filter to use here so that we can have a low phase lag feedback to the ETM?
We could try inputing a 4kHz carrier modulated width a depth of a few Hz at a modulation frequency of F1. Then we could take an FFT of the output of the discriminator and measure the width of the peak at F1 Hz. This seems like an arduous way to measure the frequency noise at a single frequency though.
It'll definitely be sensitive to DC offsets but there is already a filter bank on the INPUT filter so we can shape that as necessary. We could probably band-pass that from [4.5 - 5.3kHz] (which would correspond to a range of [73,87] MHz into a 2^14 frequency divider.