This plot shows ETM oplev and OSEM trend for 10 hours on day before yesterday as almost the same as plot shown this entry. I reported the 10-30minites fluctuations were seen, but I noticed it comes from not suspension but from oplev power fluctuation.
After Kiwamu fixed the ETM OSEM touch yesterday afternoon, still the same trend was seen, so we had thought what we fixed was not enough. This morning I looked at the yesterday's and day before yesterday's trend and noticed the simila trend both the pit and yaw in ETM oplev but not on the OSEM trend. Kiwamu suggested me to put the oplev sum on the same plot. It was!
So, ETMX is not bad, but in fact, still alignment fluctuation exist on the cavity. ITM?