Kiwamu and I roughly calibrated the analogue output from the SR620 frequency counter yesterday. The input channel, intuitively named C1:PSL-126MOPA_126MON, now reads the measured frequency in MHz with an error of about 0.1MHz - this is, I think, due to the bit noise on the D/A conversion that Kiwamu discovered earlier. That is, the output range of the SR620 corresponds to around 100MHz and is digitized at 10-bit resolution, and ...
100MHz/(10^2) ~= 0.098MHz. [Sad Face]
We set the EPICS range to [-100, 100] (corresponding to [-5V, 5V]), connected a Marconi to the Freq Counter, input a variety of different frequencies and measured the counts on the EPICS channel.
The linear fit to the calibration data was F = 2.006*EPICScount - 0.2942. From this we worked out the maximum and the minimum for the range settings that give the channel in MHz: EGUF = -200.8942 and EGUL = 200.3058. The previous range was [-410, 410]