Not logged in
Fri Mar 21 09:02:03 2008, steve, Update, VAC, tp 2 failed
Fri Mar 21 11:54:38 2008, rob, Update, VAC, tp 2 failed
Fri Mar 21 09:02:03 2008
Reply to this:
tp 2 failed
Small turbo #2 is the forepump of the maglev.
It failed last night, shut down the maglev and interlock closed V1
Ifo pressure is 20 mTorr now. The Yarm was still locked at 8am this morning.
The PSL beam to MC was blocked just before the output periscope.
The psl mechanial shutter did not work from epic screen.