We took a look-see at the PRM after the gluing from last night. The balance is still okay. The reflected beam is a teeny bit below the laser aperture (center of the beam maybe ~2mm below, so ~1mRad low). This is within our okay range, since the DC offset that the OSEMs will give will be even more, and the coils can definitely handle this kind of offset.
We took the optic out of the tower, and gave it to Bob and Daphen to bake over the weekend.