We need DAQ channels activated to measure Q-values of the ringdowns for each DOF, each optics with the Dataviewer.
What we did:
1. Activated the following DAQ using daqconfig (in /cvs/cds/rtcds/caltech/c1/scripts).
(XX=BS,ITMX,ITMY YYY=PRM,SRM ZZZ=MC1,MC2,MC3 AA=UL,UR,LR,LL,SD BBB=POS,PIT,YAW)
2. Set datarate to 2048 for each DAQ.
3. Restarted fb(frame builder).
We succeeded in making DAQ channels appear in the Dataviewer signal list, but we can't start the measurement because c1mcs is still flaky.
We found that c1mcs crashes everytime when turning off all the damping servo (using "Damp" buttons on the medm screen).
It doesn't crash when all the filters are off.