The output range of the OSEM is 0-2V.
So, the OSEM output should fluctuate around 1V.
If not, we have to modify the position of it.
What I did:
Measured current outputs of the 5 OSEMs for each 8 suspensions by reading sensor outputs(C1:SUS-XXX_YYPDMON) on medm screens.
White: OK (0.8~1.2)
Yellow: needs to be fixed
Red: BAD. definitely need fix