Because of the in-vac work on Oct. 4th (see this entry) , ITMX's OSEM offsets were changed.
The two upper OSEMs are still fine, but LL and LR seem to be out of the OSEM's range.
The plot below shows the trends of LL's and LR's readouts for about two weeks. (The channel name are in the old convention, i.e. ITMY)
Some data were missing due to the upgrade of the frame builder.
It is apparent that the offsets are changed after the in-vac work on Oct. 4th, and now they just show almost zero numbers.
The damping of ITMX can still work, if LL and LR are disabled.
At some point before pumping down, we have to check the leveling of the ITMX table again.