We found a small PCB defect which is an excess copper shorting circuit on the daughter board,
it was removed and the signal on mixer monitor path is working properly.
We were checking the new TTFSS upto test 10a on the instruction, E1000405 -V1. There was no signal at MIXER mon channel.
It turned out that U3 OpAmp on the daughter board, D040424, was not working because the circuit path for leg 15 was shorted
because of the board's defect. We can see from fig1 that the contact for the OpAmp's leg (2nd from left) touches ground.
We used a knife to scrap it out, see fig 2, and now this part is working properly.