Not logged in
Fri Sep 24 22:58:40 2010, josephb, Update, CDS, Modified front end medm screens
Sun Sep 26 18:29:23 2010, rana, John, Update, CDS, Modified front end medm screens
Sun Sep 26 18:29:23 2010
In reply to:
Modified front end medm screens
I notice on first glance:
The OSEM Sensor Input matrix and the DOF2COIL Output matrix screens should be their own screens and linked from the overview. Right now they are not.
Where is the input matrix?
The SIDE GAIN looks like zero on the main screen, but the side OSEM signal seems to be getting through to the SIDE filter bank.
. I think the wiring of the SIDE signal through the input matrix is bogus.
The OUTPUT matrix seems to be the transpose of the previous OUTPUT matrix and we have lost the wires that connect the inputs and outputs to the matrix. We ought to think about how best to represent things on the OVERVIEW screen; probably only need to have a minimal representation and allow power users to open up the detailed screen.
The TIME string is whited out. How will this be done? Does each FE display its local time on its EPICS screens?
So far unable to get any channels on DV. How do we look at channels / test points?
As far as we can tell, there is no connection between the output of the SUSPOS, etc. filter banks and the OUTPUT MATRIX. So....nothing actually goes to the coil driver. Its hard to imagine that this new SUS could have ever worked. Is there any evidence that the damping actually worked in the past, or was it something like "well, the watchdog values came down to small numbers eventually..." ???
We are trying to debug the simulink file, but....the wiki entry on how to do this is out of date (yet updated as recently as August!) some path stuff just probably needs to be edited.
Basically the suspensions are not functioning yet and we can't attempt locking of the MC.