The RefCav is locked and aligned. I changed the fast gain sign by changing the jumper setting on the TTFSS board. The RefCav visibility is 70%. The FSS loop ugf is about 80 kHz (plot attached. there is 10 dB gain in the test point path. this is why the ugf is at 10 dB when measured using in1 and in2 spigots on the front of the board.) with FSS common gain max out at 30 dB. There is about 250 mW coming out of the laser and 1 mW going to RefCav out of the back of the PMC. So the ugf can be made higher at full power. I have not made any changes to account for the PMC pole (the FSS is after the PMC now). The FSS fast gain was also maxed out at 30 dB to account for the factor of 5 smaller PZT actuation coefficient - it used to be 16 dB according to the (previous) snap shot. The RefCav TRANS PD and camera are aligned. I tuned up the phase of the error signal by putting cables in the LO and PD paths. The maximum response of the mixer output to the fast actuator sweep of the fringe was with about 2 feet of extra cable in the PD leg.
I am leaving the FSS unlocked for the night in case it will start oscillating as the phase margin is not good at this ugf.