All the necessary cables are in our hands, such as SCSIs, 37 pin D-sub and 3 pin power cables.
With a big help from Jenne and Koji, we made 37 pin F/M cables and 3 pin power cables on the last Tuesday.
Now all the cables are connected to the machines except for the 3 pin power cables.
To install the power cables I will switch off Sorensens at new 1X5 for a minute.
The model file named C1SUS was made and I succeeded in compiling and building it. So it is ready for the test.
But some medm screens look like not correctly complied.
For example the channel names which are listed on C1SUS_MONITOR_ADC0.adl aren't correctly assigned.
All the ADC channels are working. Also the channel assignments are correct.
I checked all the ADC channels if it was working while I ran the front end realtime code C1SUS.
By injecting a signal from a function generator directly to the SCSIs, I could clearly see the numbers hopping on medm screens.
None of them are working although the computer can recognize that all three DAC cards are mounted.
I think something in the IOP file and the model file may be wrong because this symptom looks similar to that of C1ISCEX we tested two weeks before ( see this entry )
I have to fix it anyhow because the DACs are very necessary part for this damping test.
They didn't work at all. Even the computer didn't recognize the binary output cards.
I think I should put some magical components on the model files.
- Binary Outputs
- simlink realtime model with the new CDS PARTS ( see the notice from Joe )
To be done