Since we now have a good measurement of the phase noise of the Rb clock Marconi locked to the Rb clock, I wanted to use that to check out the old DAQ system:
I used Megan's phase noise setup - Marconi #2 is putting out 11000013 Hz at 13 dBm into the ZP-3MH mixer. Marconi #1 is putting out 3 dBm at 11000000 Hz into the RF input.
The output goes through a 50 Ohm load and then a Mini-Circuits BNC LP filter (either 2 or 5 MHz). Then an SR560 set for low noise, G = 5, AC coupling, 1-pole LP @ 1 kHz.
This SR560 output goes into the channel C1:IOO-MC_DRUM1 (which is sampled at 16384 Hz with ICS-110B after the usual Sander Liu AA chassis containing the INA134s).
This is the 0.3 mHz BW spectrum of this test - as you can see the apparent linewidth (assuming the width is all caused by the DAQ jitter) is comparable to the BW and therefore not resolved.
Basically, the Hanning window function is not sharp enough to do this test and so I will do it offline in Matlab.