40m QIL Cryo_Lab CTN SUS_Lab TCS_Lab OMC_Lab CRIME_Lab FEA ENG_Labs OptContFac Mariner WBEEShop
 40m Log Not logged in  Sun May 2 13:04:51 2010, Koji, Summary, SUS, Coil Actuator Balancing and Spot Position  Sun May 2 15:38:12 2010, rana, Summary, SUS, Coil Actuator Balancing and Spot Position Sun May 2 16:52:44 2010, Koji, Summary, SUS, Coil Actuator Balancing and Spot Position
Message ID: 2863     Entry time: Sun May 2 13:04:51 2010     Reply to this: 2865
 Author: Koji Type: Summary Category: SUS Subject: Coil Actuator Balancing and Spot Position

I liked to know quantitatively where the spot is on a mirror.

With an interferometer and A2L scripts, one can make the balance of the coil actuators
so that the angle actuation does not couple to the longitudinal motion.
i.e. node of the rotation is on the spot

Suppose you have actuator balancing (1+α) f and (1-α) f.

=> d = 0.016 x α [m]

Full Imbalance   α = 1      -> d = 15 [mm]
10% Imbalance α = 0.1   -> d = 1.5 [mm]
1% Imbalance   α = 0.01 -> d = 0.15 [mm]

Eq of Motion:

I ω2 θ =  2 R f
(correction) - I ω2 θ =  D f cos(arctan(L/2/D))
(re-correction on Sep 26, 2017) - I ω2 θ =  D f

m ω2 x = 2 α f ,
(correction) - m ω2 x = 2 α f ,

where R is the radius of the mirror, and D is the distance of the magnets. (kinda D=sqrt(2) R)

d, position of the node distant from the center, is given by

d = x/θ = α I / (m R) = 2 α β / D,

where β is the ratio of I and m. Putting R=37.5 [mm], L=25 [mm], β = 4.04 10-4 [m2], D~R Sqrt(2)

i.e. d = 0.015 α [m]

 Attachment 1: coil_balance.png  36 kB ELOG V3.1.3-