I scanned the temperature of the crystal oven on Friday night in order that we can find the optimal temperature of the crystal for SHG.
The optimal temperature for this crystal was found to be 36.2 deg.
The crystal is on the PSL table. The incident beam on the crystal is 27.0mW with the Newport power-meter configured for 1064nm.
The outgoing beam had 26.5mW.
The outgoing beam was filtered by Y1-45S to eliminate 1064nm. According to Mott's measurements, Y1-45S has 0.5% transmission for 1064nm, while 90% transmission for 532nm. This means I still had ~100uW after the Y1-45S. This is somewhat consistent with the offset seen in the power-meter reading.
First, I scanned the temperature from 28deg to 40deg with 1deg interval.The temperature was scaned by changing the set point on the temperature controller TC-200.The measurements were done with the temperature were running. So, the crystal may have been thermally non-equilibrium.
Later, I cut the heater output so that the temperature could be falling down slowly for the finer scan. The measurement was done from 38deg to 34deg with interval of 0.1deg with the temperature running.
I clearly see the brightness of the green increase at around 36 deg. The data also shows the peak centered at 36.2deg. We also find two lobes at 30deg and 42deg. I am not sure how significant they are.