1) True. My bad. In my elog entry (but not in my fit code) I forgot the impedance Z= 750Ohm (as in the fit) of the resonant circuit in front of the square root: Vn = Vdn + Z * sqrt( 2 e Idc )
2) That is exactly the point I was raising! The measured dark noise at resonance is 2x what I expect.
I also admitted that the data points were few, especially around the 3dB point.
Today I'm going to repeat the measurement with a new setup that lets me tune the light intensity more finely.
Here's another measurement of the noise of the REFL11 PD.
This time I made the fit constraining the Dark Noise. I realized that it didn't make much sense leaving it as a free coefficient: the dark noise is what it is.
Result: the transimpedance of REFL11at 11 MHz is about 4000 Ohm.