Morning condition: vacuum rack power is still off, no MEDM screen reading.....meaning unknown vacuum pressure.We closed PSL shutter immediately.
Joe restored c1iscepis and Jenne powered up the vac-rack UPS. Now the rest of the vac-rack power were restored from starting at the top to bottom.
P1 was reading 15 mTorr. We restarted pumps and set vacuum valve positions. V1 opening required Rob's recipe of elog # 1863 to defeat interlock that
has a non communicating gauge: PTP1
CC1 pressure just reached 1e-6 Torr at VAC NORMAL configuration.