In the middle of the last month, Kiwamu and I went to Garilynn's lab to measure the phase maps of the new ITMs and SRMs.
Analysis of the phase map data were posted on the svn directory:
The screen shots and the plots were summarized in a PDF file. You can find it here:
The RoCs for all of the PRMs are turned out to be ~155m. This is out of the spec (142m+/-5m) although the actual effect is not understand well yet..
These RoCs are almost independent from the radus of the assumed gaussian beam.
In deed, I have checked the dependence of the RoC on the beam spot position, and it turned out that the RoCs vary only little.
(In the SRMU01 case, for example, it varies from 153.5m to 154.9m.)
The beam radius of 3mm was assumed. The RoCs were calculated 20x20mm region around the center of the mirror with a 2mm mesh.