This is a plot of the R and T of the existing ETM's HR coating. I have only used 1/4 wave layers (in addition to the standard 1/2 wave SiO2 cap on the top) to get the required T.
The spec is a T = 15 ppm +/- 5 ppm. The calculation gives 8 ppm which is close enough. The calculated reflectivity for 532 nm is 3%. If the ITM reflectivity is similar, the signal for the 532 nm locking of the arm would look like a Michelson using the existing optics.