The up and down scripts accessible from the OAF (still C1:ASS-TOP) screen are now totally functional and awesome. They are under the blue ! button. The up script can either be for the Seismometers, or the Accelerometers at this time. The only difference between these 2 is which burt restore file they look at: the seismometer version puts all 7 seismometer channels in the PEM selecting matrix, while the accelerometer version puts the 6 accelerometer channels in that matrix. Both scripts also turn on HP_1mHz filters in the ERR_EMPH filter bank and all of the witness filter banks, and the AA32 and AI32 filters in ERR_EMPH, CORR and PEM filter banks. This makes all of the starting filters the same between the witness paths and the error path.
If you want to use a different combination of sensors, run one of the up scripts, then change the PEM matrix by hand.
The down script disables the output to the optics, and resets the adapted filter coefficients. DO NOT use this script if you're trying to "pause" the filter to take some nice long averages.